Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Search found 1 match
- April 24th, 2017, 6:39 am
- Forum: PicoAmmeters
- Topic: how to sweep from -10V to +10V and back from +10V to -10V
- Replies: 1
- Views: 17600
I want to measure the current using 6487 Keithley sweeping the voltage from -10V to 10 V and back from 10V to -10V in multiple loops.
How can I edit the program in Keithley 6487?