Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Search found 1 match
- April 20th, 2018, 10:14 am
- Forum: Power Analyzers
- Topic: Automate PA1000 with Python
- Replies: 1
- Views: 5511
You should insert those lines in your code when working with PA1000:
analyzer.read_termination = '\n'
analyzer.write_termination = '\n'
inst.write("*IDN?"+"\n") doesn't work in this case.
NI MAX write/read the messages of that instrument with those terminations automatically (*IDN?\n).