Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have a small problem.
I have making measurement with model 2400, but everytime I get a error -113 and I don't know what I'm doing wrong.
string send to Keithley is something like this:
- Keithley Applications
- Posts: 1390
- Joined: October 15th, 2010, 10:35 am
- Country: United States
The most common cause for a -113 error is a simply syntax problem.
I ran each of your commands to a 2400 and do not see any -113 troubles.
I did experience a -221 Settings conflict when sending the :SOUR:CURR 0.001 command. This is because the instrument is on a 100uA source range and the command is asking for 1mA of source current.
Can you single step through your program and isolate which command is the offender? You probably have a missing quote or something simple.
One suggestion: first send a *RST command to put the instrument into a known state.
When the instrument is in the *RST state, source auto range is enabled. This state then allows the instrument to automatically change to an appropriate source range for the :SOUR:CURR 0.001 command.
Who is online
Users browsing this forum: No registered users and 4 guests