Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- (14.52 KiB) Downloaded 1190 times
- Keithley Applications
- Posts: 2845
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Two things were needed.
That should help.
Actually i'm using Labview 2013 hence the .VI u send is not opening in my system as of urs is in 15 version Labview . So can u send by other way so that i can visualize it. Thanks in Advance.
Sir, Thank you. Your VI working for a GPIB interface. But when I connect the instrument using serial interface(RS232-USB), It gives errors -150, -140 (character data error and string data error)and also it says "timed out error before the operation completed....". what do I miss? (P.S: I have installed the NI serial driver. When I read through Ni MAX also, it gives the timed out error)
Who is online
Users browsing this forum: No registered users and 2 guests