Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Models 2400, 2401, 2410, 2420, 2425, 2430, 2440, 6430
- Posts: 1
- Joined: March 10th, 2016, 7:28 am
- Country: United Kingdom
We have a Keithley 2400 SMU that produced a spark from a probe and now doesn’t read anything sensible, reading 23v without anything connected. It was not currently measuring or under any load at the time. It seems something has blown inside the unit however it still turns on fine without displaying any error message. I'd like to repair the unit as it still seems to function. Any ideas what I could try next for troubleshooting?
- Keithley Applications
- Posts: 107
- Joined: July 1st, 2014, 10:31 am
- Country: United States
Sounds like the instrument needs to come back for servicing.
To set up an RMA please call our service department at 503-627-2445.
Who is online
Users browsing this forum: No registered users and 2 guests