Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
The setup of the Sourcemeter :
Firmware : C33
Baudrate : 9600
Databits : 8
Parity : None
Flowcontrol : None
Terminator : <CR>
For Kickstart i’m unsing Version 22.214.171.124
The source meter runs fine when I use a serial connection and sending SCPI command’s to it. Also the source meter works fine with one of the LabVIEW example program’s for this source meter.
Ones I use the Kickstart program and use the “Refresh Instrument Discovery” Button, it start looking for all possible connected instruments on all the com port that I have. It sends out the *IDN? command to see which instruments are connected.
In my case it’s connected to Com3 (see attached file) and I can see with a monitor program that it sends out the *IDN? command and the source meter respond to it.
But at the end of the discovery action, the Kickstart program is not able to find it.
Does anybody know what I’m doing wrong. ?
Do I need to use a GPIB connection instead ?. Sounds strange, because its send out command’s on the com ports.
- Keithley Applications
- Posts: 2845
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Sorry. Kickstart does not support Serial (RS-232).
That is why you are having these issues.
I would recommend using a USB to GPIB adapter. Keithley Model KUSB-488B is a fine example.
Is there any software like kickstart or Labtracer that can communicate using RS-232 with 2400 SMU?
If not, any expected date of release?
Who is online
Users browsing this forum: No registered users and 1 guest