Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have a Keithley 2400 unit and I need to do some resistors measurements.
I already was able to communicate via rs-232 with my computer using terminal emulator.
I would like to have a simple example of the commands to send to the instrument to set it to measure the resistors using only 2 wires, these resistors are already installed in a PCB with some other digital low voltage IC´s so I would like not to damage those other IC´s that is why I only want to activate resistor measurement.
My procedure will be to connect the 2 leads to the resistors and send the command and read the resistor value back to the terminal emulator, then move the leads to the other resistor and read and so on.
- Keithley Applications
- Posts: 402
- Joined: June 10th, 2013, 6:00 am
- Country: United States
The above script uses 4-wire so you will want to change the sense mode
You may want to change the range for your particular application. If your resistance is below, say 100 ohms, you will want to use 4-wire which is also called remote sense. Offset compensation may be called for depending on what accuracy you require if you are measuring low ohms.
Who is online
Users browsing this forum: No registered users and 3 guests