Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Models 2400, 2401, 2410, 2420, 2425, 2430, 2440, 6430
- Posts: 18
- Joined: May 15th, 2013, 6:02 am
- Country: China
In the 2400 User manual P4-11, there is the description on " Ohms source readback selection" with front panel operation. However, I cannot find how to remote setting "Ohms source readback". Any solution? Thanks.
PS: the manual P4-11 description--
Ohms source readback selection
Use the following procedure to enable or disable ohms source readback:
1. Press CONFIG then Ω.
2. Select SRC RDBK, then press ENTER.
3. Select DISABLE or ENABLE as desired, then press ENTER.
4. Press EXIT to return to normal display.
- Tektronix Applications
- Posts: 266
- Joined: April 26th, 2010, 12:01 pm
- Country: United States
The instructions you listed from the manual are correct. However, what they omit is that after pressing CONFIG and then pressing Ohm you need to press the right arrow key several times to get to the page where you can select SRC RDBK. Keep pressing the right arrow. It's the last item in the menu.
Who is online
Users browsing this forum: No registered users and 4 guests