Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I decomposed the 'Sweep and Acquire Measurement' VI and went down to SCPI command level, using this command ":SOUR:SWE:LIN." However, if I naively give the command with different starting/stop level in series, only the last command's data gets stored in the buffer. Can someone please help me out how to perform a sequence of linear sweeps without delays in-between?
Lastly, I then tried to use 'Read Multiple' VI, in which I supposedly give a list of voltage levels I want the sourcemeter to sweep across. Generating a list of numbers is certainly doable in LabVIEW, but I just don't understand this VI (attached screenshot). Where do I actually put the list? Is it the "manual range"? Why is it just one number then?
I'd greatly appreciate any help. Thank you in advance!
- Screen Shot 2018-06-10 at 1.18.14 AM.png (229.82 KiB) Viewed 2037 times
Who is online
Users browsing this forum: No registered users and 3 guests