Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I inherited some LabVIEW SW that Reads the Voltage on a UUT.
When the 2410 is switched in (Relays) the UUT output is loaded down.
I put a 1.5 Battery in series with a another external Current meter, during measurement it draws 200mA.
I have attached an image of the VI.
What I am not sure about is what was in the original meter's *RCL 0.
I have another post on that topic.
Thank you, Jeff
- 2410 Read Voltage.png (48.26 KiB) Viewed 193 times
- Keithley Applications
- Posts: 1262
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Who is online
Users browsing this forum: No registered users and 3 guests