I inherited some LabVIEW SW that Reads the Voltage on a UUT.
When the 2410 is switched in (Relays) the UUT output is loaded down.
I put a 1.5 Battery in series with a another external Current meter, during measurement it draws 200mA.
I have attached an image of the VI.
What I am not sure about is what was in the original meter's *RCL 0.
I have another post on that topic.
Thank you, Jeff
- 2410 Read Voltage.png (48.26 KiB) Viewed 3658 times
- Keithley Applications
- Posts: 1427
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Who is online
Users browsing this forum: No registered users and 1 guest