Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I use Keithley 2410 source-meter to do TDDB measurement. I want this source-meter stop at current compliance.
I have tried to put this code
but it didnot work. Any help would be greatly appreciated.
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
If so then use the compliance abort command.
:SOURCE:SWEEP:CABORT (NEVER,EARLY or LATE).
If not using a sweep then you could also use the trip command.
This would return a yes(1) or no(0) when addressed to talk.
Who is online
Users browsing this forum: No registered users and 3 guests