Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am trying to program the 2400 to be able to sweep voltage at specific steps ( a list of 10 values for instance). I need to be able to present each value after a trigger event via the tlink #1. The trigger events might not occur at a fixed rate. I have multiple 2400 that will receive this signal from a pattern generator. Is there some SCPI code anyone can share that can give me a starting point?
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Each of these can output a pulse or none of them. But only one at time.
The commands are as follows...
Do this in a LIST sweep.
That should output a pulse at each step on the SENSE(Measurement) phase.
Who is online
Users browsing this forum: No registered users and 4 guests