Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I was making a measurement using LV and Keithley 2400 with serial comunication.
During measurement where I have set moving filtering to 20 and speed on 10NPLC, after a while I got a error -363 Input buffer overrun.
Could anyone help me to avoid this error or at least how to clush buffer in case of this kind of events?
I was checking datasheets but I haven't find any clues what to do.
At the moment only thing that set Keithley back to it's feet is a manual Off/On.
Since I'm making a industrial measurement program, this is not a option
Thank You for help
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
This helps with long commands and large data transfers.
Who is online
Users browsing this forum: No registered users and 1 guest