Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am currently controlling a Keithley 2425 using SCPI commands through ATEasy. I tried using the command sequence for source-measure example in page 1-15 of the Quick Start Guide. Every time I send the command sequence with an increment in voltage level, the output toggles OFF and ON before the next voltage level is set. What command sequence can I use in order to not have the output toggle OFF and instead staircase to the next voltage.
I am aware of the Sweep function, but I need other to designate other equipment to do other functions while the SMU is still at a specific voltage level. Put simply, I just really need to avoid to avoid the output turning OFF when changing the voltage level. Essentially I want to mimic what happens when the SMU is controlled manually with the output ON and one presses the UP and DOWN arrow keys to adjust the voltage level.
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Now read the data
Notice the unit goes to 9V
Notice the unit goes to 8V.
And so on...
Who is online
Users browsing this forum: No registered users and 3 guests