Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I met one problem on the output current. when we source 28V on our EVB and measure the current, the max current was Clamped at 1049.9mA when our real current should be about 1.6A. so our sample didn't work normally. could we change the current measurement range to a higher level. we noticed that the biggest clamp current on 2430 was 3A, how could we get our EVB work normally using Source Meter 2430?
- Keithley Applications
- Posts: 254
- Joined: February 28th, 2011, 6:17 am
- Country: United States
The data sheet available for download here can explain this graphically: http://www.keithley.com/support/data?asset=372
Page 2 of that PDF has the graphs that represent the power envelopes.
Who is online
Users browsing this forum: No registered users and 2 guests