Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have been able to get the trigger part of it programmed correctly but I am having trouble getting the meter to source correct voltage before trigger, during SDM and after SDM. Do I have to program a sweep here?
- Keithley Applications
- Posts: 254
- Joined: February 28th, 2011, 6:17 am
- Country: United States
Hope this helps. Also, just in case you did not know, the output off state of the 2400 is customizable to a high-impedance mode where a relay disconnects the jacks from the analog circuitry inside for theoretically a very high impedance.
Who is online
Users browsing this forum: No registered users and 6 guests