Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
We have a Keithley 2440 source meter. If the unit is on and warmed up, but the leads are off (the unit is idle and is not applying power or sensing), how long can the unit be left on without needing to worry about a loss of accuracy or incurring wear.
Obviously the unit should be shut-down whenever possible, but we are curious about this idle-state.
Thanks, I know it's an unusual question.
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
This unit, the 2440 should be the same.
They are designed to operate for one year until the one year calibration cycle is passed.
Even then it will still operate, just not be in calibration.
But you could leave it powered on and in "idle" indefinitely.
Who is online
Users browsing this forum: No registered users and 3 guests