Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Models 2601A, 2602A, 2611A, 2612A, 2635A, 2636A
Models 2601B, 2602B, 2604B, 2611B, 2612B, 2614B, 2634B, 2635B, 2636B
Im rather new to this, so sorry if I am asking a silly question.
I want to measure pulse voltage using the function PulseIMeasureV. I have a 2601B SMU which I am controling with Keithley 2600 Series Labview driver. Printscreen is included in attachments (is potato quality, so either download or try the VI).
1) Im getting the error code 5038: Index Exceeds Maximum Reading and my buffer data are: 9.91000e+37.
2) There is no buffer connection in ConfigPulseIMeasureV. It is in ConfigPulseVMeasureI, but Im getting the same error as in 1).
I guess Im doing something wrong with buffers?
Thanks for all replies!
- (34.81 KiB) Downloaded 12 times
- PulseIMeasureV.jpg (162.63 KiB) Viewed 149 times
I don't know anything about Labview, I always used TSP scripts from a flash drive.
I had a similar problem before, and my solution was to get the factory scripts and modify them. You could look at the reference manual for "Retrieving and modifying a factory script listing" (page 9-20 in the Rev.E January 2019 edition).
The buffer inside one of these factory scripts was a non-volatile buffer, which are smaller. I changed that to a buffer whose size I could define at runtime.
Hope it helps!
Who is online
Users browsing this forum: No registered users and 1 guest