Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Models 2601A, 2602A, 2611A, 2612A, 2635A, 2636A
Models 2601B, 2602B, 2604B, 2611B, 2612B, 2614B, 2634B, 2635B, 2636B
Channel A sources a negative 500µs current pulse. The pulse level, starting at -600µA is increased to -850µA in 1µA steps. The interval between 1µA steps is 100ms. The source changing amplitude is displayed on the bottom left on the SrcA field.
Channel B sources -10V to the collector with the compliance level set to -150mA. How can the pulsed collector current be displayed during this operation?
i would appreciate your feedback and suggestions.
Who is online
Users browsing this forum: No registered users and 1 guest