Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Models 2601A, 2602A, 2611A, 2612A, 2635A, 2636A
Models 2601B, 2602B, 2604B, 2611B, 2612B, 2614B, 2634B, 2635B, 2636B
I have error code 5001/5003 and 5051 on 2 different power supplies (refurbished equipement received in December).
I already had this issue end of last year on a third Keithley used on the same test bench (automatic prober for 200mm LED wafers/ IVsweep performed). This Keithley was send to service.
I'm worried and i would like to understand what could be the root cause of such issue in order to avoid same problem in the future...
Thanks in advnace for your support.
Who is online
Users browsing this forum: No registered users and 3 guests