Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Models 2601A, 2602A, 2611A, 2612A, 2635A, 2636A
Models 2601B, 2602B, 2604B, 2611B, 2612B, 2614B, 2634B, 2635B, 2636B
I am an amateur in terms of TSB. The SMU I am using is a 2612b.
I am trying to set up a script to measure Ids and Ig under pulsed gate voltage (Vg) for an OFET, while maintaining a constant drain voltage (Vds) during the measurement. The pulsed Vg can be alternating from -1V to 1V and Vds is kept at -1V.
I can call PulsedSweepVSingle(-1, -1, 10, 0.1, 0.2, 0.1, 1, false) and set smua.source.levelv = 1 V as shown in the attached text. This would faithfully output a 20 pulsed Vg at smua alternating from -1 to 1 V as checked by oscilloscope.
1) Is it possible to introduce smua.measure.iv(smub.nvbuffer1, smub.nvbuffer2), so that not only i and v of smua can be measured, but i, v of smub can also be measured simutaneously, with a common timestamps?
2)The scripts now idle/stop i measurement after the pulsed Vg return to its off value 1 V, is it possible to keep measuring i even after Vg is in its off value?
3) The i measurement only returns single value at the end of each pulse, it there a way to output designated number of data points? Say maybe 10 or 20 points of i data regardless of on and off states.
Thanks for helping me on this issue.
- KE26XXB Pulsed sweep single 1.txt
- (3.33 KiB) Downloaded 15 times
Who is online
Users browsing this forum: No registered users and 3 guests