Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Models 2601A, 2602A, 2611A, 2612A, 2635A, 2636A
Models 2601B, 2602B, 2604B, 2611B, 2612B, 2614B, 2634B, 2635B, 2636B
I think that the issue is more of a set-up issue than the measurement itself. Since you have a 2636A, you can actually use the TSP Express to do your specific measurement and settings. Another aspect of the TSP Express is that it can generate the code for you as well. Use this as your debugging tool to make sure that you achieved your measurement needs. Also make sure that you are actually connected properly (HI = RED on the triax alligator, LO = RED on the triax alligator).
See link below;
http://www.keithley.com/products/dcac/c ... B/tsp.html
Who is online
Users browsing this forum: No registered users and 4 guests