Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I would like to do the following measurement:
- Apply with a different component a arbitrary wave with a frequency around kHz.
- Measure with the Keithley 6430 current and voltage, if possible, or just current in that frequency.
I was trying to accomplish that using LabVIEW but I can't measure in any situation the voltage and current simultaneously in the two-wire sense connection. Can you help me giving me some advices of how could I measure in such way?
When I press toggle it measures the current correctly but the voltage is always zero (voltage source with 000.00mv range)
- Keithley Applications
- Posts: 1245
- Joined: October 15th, 2010, 10:35 am
- Country: United States
A SourceMeter can be configured to behave like an current meter by configuring it to force 0V and measure I. Wire it in series with the device.
It can be a volt meter by configuring it for force 0Amps and measure V. Wire it in parallel with the device.
Overall reading rates for the 6430 will be much slower than the 1KHz. Check out pg 4 of the datasheet.
https://www.tek.com/search?keywords=643 ... Adatasheet
Who is online
Users browsing this forum: No registered users and 1 guest