Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Posts: 1
- Joined: February 12th, 2018, 11:22 am
- Country: United States
On all of our 6517A's the sign bit's value is always "-" when the meter is on the 10 nano amp scale and the response format is set to "SRE" or "REAL,32". However the faceplate indication always indicates the correct polarity for the input signal and if the meter's format is set "ASC" the response is always correct as well. Is this a known bug with the old 6517A's and is there a fix for it? All of our meters have a firmware revision of C05 A02.
Who is online
Users browsing this forum: No registered users and 3 guests