Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
For my experiments, I use a self-written LabVIEW code that writes commands to and reads output from a Keithley 6517B electrometer, through GPIB connection. The problem is when I try to implement zero correction on the charge; the LabVIEW code does not return any errors, but the display of the electrometer gives 'error code -200' (execution error). The chain of commands in which this error occurs is as follows:
Apart from the error code, the code works fine. As suggested in this topic: viewtopic.php?t=14720, I tried to put some time delays between the commands, however to no avail. Does someone know a solution for this problem?
Thank you in advance!
Who is online
Users browsing this forum: No registered users and 8 guests