Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
If I disconnect the photodiode cable, the 2701 will fluctuate within the range of -100 to -40 mV. Is this normal?
- Keithley Applications
- Posts: 1341
- Joined: October 15th, 2010, 10:35 am
- Country: United States
If front terminals, make sure the front/rear switch selector is set to front. Use rear for a channel from a 77xx switch card.
When a voltage measuring device is open circuit, the reported voltage will be randomly anything. Try touching HI to LO and see if voltage goes to near zero.
As for measuring voltage from the photodiode, usually a voltage bias is applied and the current is measured and correlated to the photons. Or does your device have some amplifier output that gives you a voltage on BNC?
Who is online
Users browsing this forum: No registered users and 1 guest