Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
After running speed diagnostics to determine where the program was getting hung up, it seems that retrieving data from the buffer takes a large amount of time. Even when the buffer size is set to two, the command :TRAC:DATA?; takes approximately 700 ms to run. I checked and the data packet is only 58 bytes. It does not seem to me that it should take so long for a packet that size to transfer from the instrument to the computer. This limiting my sampling speed to about once a second but I would appreciate if I could sample 5-10 times a second. My experiments last several hours so it is not possible to simply store it all in the buffer and retrieve the data at the end.
I am new to SCPI programming as well as buffer storage and have been attempting to read and understand using the 2182A manual and various - sources. I am struggling to diagnose what is causing this program to run so slowly.
Who is online
Users browsing this forum: No registered users and 1 guest