Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 1244
- Joined: October 15th, 2010, 10:35 am
- Country: United States
You mention it wanders up and down with time. Is auto zero in use?
By how much is it changing? For the range in use, is it within the error budget? The specs are when using SLOW and a sine wave signal type.
What sort of crest factor does your signal have? Check out the AC Voltage measurements and crest factor section in the 2701 manual.
Who is online
Users browsing this forum: No registered users and 4 guests