Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
i am using a 2-channel 2612B and i want to source constant currents and measure the voltage over time as a pre-test on a LED.
Therefore i want to use one channel (smua) for sourcing and the other one (smub) for measuring.
I tried a lot, but it is not possible to do so.
Just to check each channel's ability, i used smua and afterwards also smub as source and measure channel.
-> It worked!
Why am i not using this Setup for the future?
-> In the next step i have to source again my LED, but the measurement is in another place in a circuit.
What can i do to solve the Problem?
- Keithley Applications
- Posts: 1436
- Joined: October 15th, 2010, 10:35 am
- Country: United States
To have a SMU channel behave as a voltmeter, configure that channel as a zero amp current source with a voltage compliance higher than the voltage signal you will be measuring. Use the lowest current range to force the 0 Amps.
This config will mimic a high input R voltmeter with little or no current flowing into the SMU and will minimize loading for the voltage measurement.
Who is online
Users browsing this forum: No registered users and 4 guests