Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I may have some problems with the Labview drivers or configuration. Could somebody help me ?
I need to measure a sensor signal (+10v to -10v) with a Kusb 3108 and a Labview application
When the sensor is in repose the output signal is between 1.72 V and 1.74v.
I get the same values if I measure with a voltimeter or with the "Quick data Acq" software
but if I use LabView 9 and "AinSingle.vi" example (from LV-Link) the output signal is between 7.9 V and 8.1v.
What is wrong in my measurements ??
Who is online
Users browsing this forum: No registered users and 4 guests