Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I'm interested in plotting I-V curve of a Power MOSFET. I want to do this at specific gate voltages to make a characteristic curve.
Do I need an external gate trigger or voltage source?
I did not find this information in the manuals. Maybe you can direct me to the right page.
Thank you in advance,
- Keithley Applications
- Posts: 402
- Joined: June 10th, 2013, 6:00 am
- Country: United States
Do you want to do this without spending money?
For automatic control, a Keithley 2601B would be connected to the gate. You would use TSP Express to set up the gate steps and drain sweep. Here is a video of the 2651A being controlled by TSP Express:
http://www.keithley.com/centralized_dis ... etid=55810
http://www.keithley.com/centralized_dis ... etid=57015
The demonstrations are for 2-terminal devices. To test your 3-terminal device, you configure TSP Express for a two SMU step and sweep family of curves. It is easy and quick to set up.
If you do not want to spend money, you would need to manually set the gate voltage and then run a TSP Express sweep to plot the curve.
Who is online
Users browsing this forum: No registered users and 1 guest