Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am trying to develop a battery characterization program using a 2651a and LabView. I read through Application Note 3221 as kind of a baseline and have gone forward with some programming and testing based on that paper. I am a bit confused though. During charging, I set the source to above the battery voltage (12.6V on a 3-cell Li-Ion pack) and compliance limit for the current to a set amount and the battery charges at that current until it hits the voltage and my code shuts it down. The problem lies in discharging at a set current. I set the source voltage to below the battery level (9V) and the compliance limit to a set value (say 1.5A), but that compliance limit doesn't seem to be behaving the way I would expect. The current varies approximately linearly with the limit value, but the absolute value of the incoming current is always larger than the limit. For example, with 1.5A my current is reading approximately -2.2A, if I take the current up to 2A the reading is about -2.6 or so. Am I misunderstanding something, or how do I limit this current value?
So here are my settings for "charging":
Battery = 3-cell LiIon
Source Voltage = +12.6V
Limit Current = 3.5A
Hi-Z off mode (to keep battery from draining)
Under these settings the current measured at 3.4999A and the voltage was rising.
For discharging, my settings were:
Source Voltage = +9.0V
Limit Current = 1.0A
Hi-Z off mode
Under these setting the voltage was dropping like I would expect, but the measured current was -1.13A. As I raise the limit current, the measured current dropped (further negative) somewhat linearly.
So I guess the crux of the question is, is there a way to set the Limit to a negative value? Is this even the issue or is there a setting that I am missing that will limit this value to a tighter tolerance?
- Keithley Applications
- Posts: 1281
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Front panel: Config SRC: V-Source: Sink-Mode: Enable
Remote: smua.source.sink = smua.ENABLE
Who is online
Users browsing this forum: No registered users and 4 guests