Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I have evaluating use the 2651A device for a application of generation and measurement of power AC signal with following features:
Voltage Range: 0-10 V
Current range: 0-20A
Frecuency range: 0 - 1 MHz
I need support for this device
- Keithley Applications
- Posts: 212
- Joined: October 25th, 2010, 1:31 pm
- Country: United States
The 2651A is a High Current DC Sourcemeter. It is capable of pulsed high current however it is not capable of generating a sine wave. There is one exception using a list operation to simulate a sine wave where each point in the list would be an equivalent to a point in the sine wave. The limitation of the list operation simulation is that the rate will only be about 60 Hz with reasonable resoltuion.
Who is online
Users browsing this forum: No registered users and 4 guests