Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am experiencing an error in 2651A today when doing I-V measurements, sourcing the instrument in costant current mode set at 1A.
After few seconds, the current started to increase and suddently there was no more response at all when I pressed any button. So I switched it off and on again and it shows now two errors that the manual classifies as "FATAL". They are: ERROR 5051 and ERROR 5001.
Could you suggest any solution?
Thanks a lot for your help.
Who is online
Users browsing this forum: No registered users and 4 guests