Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
However, I don't know, how to call the 2657A in ctm? Is this possible and am I able to do sampling, or is this exclusively available with TSP? Would there be another way to use a program for both systems?
- Keithley Applications
- Posts: 1323
- Joined: October 15th, 2010, 10:35 am
- Country: United States
The CTM of the ACS software is calling a user library function that is written with the KULT tool on the 4200-SCS.
The KULT tool gives access to GPIB read and write commands, so you can certainly connect the 2657A to the GPIB bus of the 4200-SCS and then write a user library that interacts with the 2657A and returns data back to the sheet.
If ACS is the application that will receive the data, use the ACSPostData() commands. If KITE is the application that will receive the data, use the PostData() commands. You could even define an input parameter to indicate which is the target application and conditionally use ACSPost() or Post() commands.
Question: will the 2657A be used alone or do you need coordinated use of the native 4200-SMU from the same user library? What kind of test routine are you trying to carry out?
Who is online
Users browsing this forum: No registered users and 2 guests