Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Does anyone know if the 2657A can provide this type of test? Will it allow us to choose a set current and automatically control the voltage in order to maintain that current?
- Keithley Applications
- Posts: 1262
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Yes, the 2657A can do that too.
All of the Keithley SourceMeter products have the feature of being able to source voltage or source current.
When asked to source current, you set the voltage compliance limit. So long as the DUT voltage remains below this voltage compliance limit, the SMU will maintain the desired source current value.
Are you expecting for the DUT voltage to be more than 1100 volts?
Who is online
Users browsing this forum: No registered users and 3 guests