Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
We are trying to carry out pulsed IV measurements using ITM on a two-terminal device. However, no data are reported from the test (in the sheet or the graph), as if no measurements were executed at all.
The test runs quickly (1-2 seconds). It reports no error. We tried connecting DUT via 1-channel and 2 channel setups (of the same PMU). No success in either setup.
Any idea about possible cause?
- pulse-problem2.png (129.68 KiB) Viewed 177 times
- pulse-problem1.png (128.52 KiB) Viewed 177 times
- Keithley Applications
- Posts: 1625
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Close KITE and invoke KCON. Run the selftest for the 4225-PMU. Any errors?
I am suspecting some sort of hardware malfunction.
Who is online
Users browsing this forum: No registered users and 2 guests