Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Posts: 1
- Joined: October 9th, 2018, 11:36 am
- Country: United States
I am trying to write a routine to apply a bias from an SMU and at the same time pulses from an HP8110. On the scope I see the SMU bias being applied and then it stops and the pulses begin. How can I keep the SMU bias going until the pulses stop? I put a delay command but it doesn't seem to cause enough of a delay to keep the smu from stopping.
Is there any KULT manual that describes the various commands in detail?
Who is online
Users browsing this forum: No registered users and 3 guests