Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
- Keithley Applications
- Posts: 1543
- Joined: October 15th, 2010, 10:35 am
- Country: United States
If you intend to use the remote sense lines (four wire connections), then the contact check fail issue should be resolved by correcting the connections.
If you do not need to use the remote sense lines, then to operate the SMU in two-wire mode, set the Sense Mode at device level to Local. When it is set to default, it makes use of the settings in the Tools>Preferences (Misc tab).
Who is online
Users browsing this forum: No registered users and 1 guest