Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
The 6517A electrometer can choose to use REL from current mode when measuring R, however I did not find this configuration on 2450. How to do it?
Btw, why my 2450 making click sound even when it's turned off? Sounds like a relay switching every few minutes.
Who is online
Users browsing this forum: No registered users and 1 guest