Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
than the previous one, there is a time delay. If both voltages are in the same
range, there is no delay.
Single range option:
The first voltage is 0.3V, the second is 1V
Dual range option:
The first voltage is 0.3V, the second is 3V
In both cases, the duration of the second pulse is 100 ms.
The dual range option is required to be the same as the single voltage range option.
How to solve this problem?
- Dual_range_option.png (1.31 MiB) Viewed 143 times
- Single_range_option.png (1.28 MiB) Viewed 143 times
- Keithley Applications
- Posts: 1562
- Joined: October 15th, 2010, 10:35 am
- Country: United States
The config list can specify not only the source value, but the range to use, the limit to apply, etc.
Additionally, a timer object can be used to generate events at well controlled intervals. A wait block in the trigger model can detect the timer events and afterward load a value from the source list.
Here is scope shot using a three point source list and 100msec time intervals.
CH2 of of the scope are digital output timing marks at DIO line 1 in response to each timer event. In attached TSP file are the lines of TSP command set code to make it happen.
- (3.24 KiB) Not downloaded yet
Who is online
Users browsing this forum: Bing [Bot] and 2 guests