Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
1:in the 2510 user manual page 56 the scale factor for the VSS is in K/A(it’s not A but V) but the problem is on the front panel on the 2510 the scale factor is in V/K, does anyone know if we must used the factor in K/V or V/K, ( the offset factor in the formule is in K, it seems normal that the scale factor is K/V). Where is the mistake on the manual or on the front panel?
2:I didn’t find in the manual the speed of the loop for a VSS sensor, the customer (CEA ) want to add between the sensor and the 2510 a filter and he needs the know the speed of the measure.
Thanks in advance for your help.
- Keithley Applications
- Posts: 2842
- Joined: June 10th, 2010, 6:22 am
- Country: United States
VSS = V / K
ISS = I/K
2. The speed is set to 16.67 msec no matter what.
This is a bug and a noise issue if you are using 50Hz power.
I will enter a problem report (NCM).
Who is online
Users browsing this forum: No registered users and 3 guests