Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am working on our Keithly 6487 - that has failed. The OCOMP indicator is on and I get the accompanying CMPL indicator. I have removed all cables. I measured -3.47Vdc at power up on the V-source output terminals and around 400mV when I change currentl limit.
After Googling the problem it seems to be quite common...
I believe if I had the power board schematic I would stand a chance of troubleshooting the problem and repairing the unit but so far I have not been able to locate it.
- Keithley Applications
- Posts: 2845
- Joined: June 10th, 2010, 6:22 am
- Country: United States
Because of IP issues Keithley does not give out the schematics of the instrumentation.
I would recommend you contact the service department.
Who is online
Users browsing this forum: No registered users and 4 guests