Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I'm using 50 Ohms for internal impedance on the 3402 and the 50 Ohms load Impedance configuration.
When I use one osciloscope with 50 Ohms input impedance the voltage reading is always double of the setup voltage on the pulse generator.
Another osciloscope gives nearly 40x the setup voltage.
What am I doing wrong?
Thanks in advance
- Keithley Applications
- Posts: 2847
- Joined: June 10th, 2010, 6:22 am
- Country: United States
When High impedence is selecte this would double the output.
Check to see which it is set to.
Who is online
Users browsing this forum: No registered users and 1 guest