Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
Many thanks Paul
Not sure about a command to confirm that, but when you configure the digitize function to wait on an edge trigger, the trigger model already has a notification block at the end of it that is configured to output a trigger on the External Out line. You will need a scope or something to probe this line and view the trigger pulse.
Another thing you could do is modify the default trigger model and add an Action block with the Log Event action after the Wait block. Now after the instrument is triggered an entry will be added to the event log thus confirming that it was successfully triggered. This is just one of many ways you could modify the trigger model to give you some indicator that the trigger was successfully received.
Who is online
Users browsing this forum: No registered users and 1 guest