Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I found when I turn 'Auto Zero' off, the reading speed works correctly. Unfortunately it defaults to on. Is there a way to correct this? Or turn it off via SCPI?
I'm trying to write a python script with PYVISA to control the device and take measurements. It would also be helpful if anyone knows of a SCPI reference for this device?
- Keithley Applications
- Posts: 1281
- Joined: October 15th, 2010, 10:35 am
- Country: United States
Code: Select all
Both command sets are documented in the Reference Guide.
Can you clarify about which A/D converter you are using? Are you using DCV or Digitize Volts measure function?
The digitize Volts can be run from 1KHz to 1MHz sample rates. But auto zero is not a feature for this mode as it would clearly introduce time gaps in the sampling.
Who is online
Users browsing this forum: Kiriakos and 1 guest