Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I need to trigger the multimeter (DMM7510), but it does not respond to the command from the other instrument (Keysight 33220A). I use the DIN to BNC trigger cable (Model 8503) to connect the equipment.
Do you have any idea of what could be the problem?
- Keithley Applications
- Posts: 1481
- Joined: October 15th, 2010, 10:35 am
- Country: United States
How are you setting up the DMM to use the trigger source?
Who is online
Users browsing this forum: No registered users and 1 guest