Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
-. Firmware Revision : C02
-. Display Revision : A02
First, setting up the continuous initialization "OFF" by command ":INIT:CONT OFF".
And next, sending the command for measurement ":READ?".
That time, changed the continuous initialization "ON" automatically.
How can I read the measurement value that do not initialization continuous on?
- Keithley Applications
- Posts: 2849
- Joined: June 10th, 2010, 6:22 am
- Country: United States
That disables continuous initiation.
Now, when sending a :READ?
It will drop down out of IDLE state and take one reading.
Then it will return to IDLE state.
Is that what you want?
Who is online
Users browsing this forum: No registered users and 5 guests