Tektronix Technical Forums are maintained by community involvement. Feel free to post questions or respond to questions by other members. Should you require a time-sensitive answer, please contact your local Tektronix support center here.
I am using 181 Nanovolmeter.
How can I get the address of this instrument.
Ni max doesn't showing any device.
I am using LabView to read data.
What are the command parameters to read voltage from it in Labview.
- Keithley Applications
- Posts: 2848
- Joined: June 10th, 2010, 6:22 am
- Country: United States
The address is selected by the rear panel address switched.
They are only read upon power up. So if you change the address then you need to recycle power to the unit to refresh the change.
This is shown on page 4-3 of the user manual.
here is the link to the manual on the KI site.
Who is online
Users browsing this forum: No registered users and 2 guests